Figure 220.127.116.11 Select test category - irradiance
Set the number of scans and integration time in [Settings], place the ground object spectrometer flat on the horizontal plane, click [Angle Correction], and correct the current angle to 0°, click [Save], the system prompts "Click to save the parameters", as in As shown in Figure 18.104.22.168, click [OK] to save the settings.
Figure 22.214.171.124 Set and save parameters
Click [Dark Current], the system prompts information as shown in Figure 126.96.36.199, click [OK] to collect dark current, and the dark current collection is completed (after dark current collection, it will be saved to the instrument, if the user does not collect dark current again, it will The last dark current is used by default), as shown in Figure 188.8.131.52.
Figure 184.108.40.206 Acquisition dark current prompt
Figure 220.127.116.11 Dark current acquisition completed
Install the cosine corrector, click [collect data] or short press the physical collection button, the system starts to collect data continuously, click [save data], as shown in Figure 18.104.22.168, enter the file name and click [OK] to save the current latest A piece of data, click [Cancel] to cancel saving.
Figure 22.214.171.124 Collection completed and data saved
If the collection setting is set to multiple collections, the system will save all the data collected this time, and the saved data file names will increase one by one from 1 to the last one according to the input file name (for example, if 3 pieces of data are collected continuously, the input file name is B , the file name is automatically saved as B-1 to B-3), and the saved data can be viewed in the history record, as shown in Figure 126.96.36.199.
Figure 188.8.131.52 History